Int Poster J Dent Oral Med 4 (2002), No. 3 15. Sep. 2002
Int Poster J Dent Oral Med 2002, Vol 4 No 3, Poster 133
Determining the Wear Resistance of Occlusal Splints in a Prospective Clinical Study
Authors: Dr. med. dent. Peter Ottl, Dentist Petra Schmelz, Dr. med. dent. Andree Piwowarczyk, Prof. Dr. med. dent. Hans-Christoph Lauer
Department of Prosthodontics, School of Dentistry, ZZMK (Carolinum), J. W. Goethe University, Frankfurt, Germany
May 17-19, 2001
50th Annual Meeting of the "Deutsche Gesellschaft für Zahnärztliche Prothetik und Werkstoffkunde"
Bad Homburg, Germany
Best Poster of Conference
To determine quantitatively the wear resistance of a newly developed light-curing splint resin over a period in situ of six months.
Materials and Methods
n=20 consecutive patients (mean age: 34.7 years; 12 F, 8 M)
- Inclusion criteria:
- Natural dentition/fixed denture
- Complete dentition to at least the 1st molar
for the stabilization splint sample:
- Insufficient occlusal support
- Increased occlusal loss of dental hard tissue
for the distraction splint sample:
- TMJ pain and
- Complete anterior dislocation of the disk without reduction/with terminal reduction
- TMJ osteoarthrosis
|Fig. 1: Stabilization splint in situ
- Resin splint material (Fig. 1):
- Light-curing (400-500 nm) resin made of high-molecular dimethacrylates with organic and inorganic fillers
- Does not contain methyl methacrylate
- Study design:
- Duration: 6 months
- Types of splints (maxilla, n=10 each):
stabilization splints, distraction splints
- Splint wear mode: 24 hours
before insertion (BI), at 4 weeks (4W),
at 3 months (3M), at 6 months (6M)
- Occlusal adjustments were restricted to the time before 4W.
|Fig. 2: Test setup
- Measuring Technology (Fig. 2)
- Vibration-isolated table framework
- 3 translation stages (for directions x, y, and z) (DC-Motor) (PI, Waldbronn)
- DV 4 stereomicroscope (Zeiss, Oberkochen)
- WA 20 inductive displacement transducer/ Spider8 digital 8-channel measurement unit/ Catman 32 software V2.1 (HBM, Darmstadt)
- Local coordinate storage for occlusal contacts during baseline measurements
- Ten measurements each in regions 13, 23, 16, 26 (BI, 4W, 3M, 6M)
- Splint repositioned on remount cast
- The medians of the occlusal vertical gains/losses (wear, resin torsion, water sorption, etc.) are shown in Fig. 3 (stabilization splints) and Fig. 4 (distraction splints).
|Fig. 3: Occlusal vertical gains/losses (medians) of the resin in vivo over a period in situ of six months (n=10 stabilization splints)
|Fig. 4: Occlusal vertical gains/losses (medians) of the resin in vivo over a period in situ of six months (n=10 distraction splints)
- Statistical analysis (Mann-Whitney U-test, p < 0.05) showed no significant differences when comparing the corresponding results of stabilization and distraction splints.
|Figs. 5a and b: Sagittal oblique images (MRI) of the condyle/fossa relationship without splint (Fig. 5a) and with distraction splint inserted (Fig. 5b) following six months of wearing
- The present study clinically confirms the good wear resistance results of the new resin splint material obtained in a previous in-vitro study [OTTL et al., Dtsch Zahnärztl Z 52, 342 (1997)].
- Good wear resistance is of great importance for maintaining the therapeutic mandibular position during the treatment period (Figs. 5a and b).
This poster was submitted by
Dr. Peter Ottl.
Dr. Peter Ottl
Department of Prosthodontics
J. W. Goethe University